Standardization of the Electrical Impedance Myography (EIM) and Detection of Breast Cancer Using EIM
Term of Award
Master of Science in Applied Engineering (M.S.A.E.)
Document Type and Release Option
Thesis (restricted to Georgia Southern)
Copyright Statement / License for Reuse
This work is licensed under a Creative Commons Attribution 4.0 License.
Department of Electrical Engineering
Committee Member 1
Committee Member 2
Adel El Shahat
Electrical Impedance Myography (EIM) is a painless non-invasive electrical technique for the assessment of neurological disease status. In this electrophysiological technique, the EIM parameters namely resistance, reactance and phase depend on several anatomic factors such as muscle girth, skin thickness, fat thickness etc. EIM may also be affected by several non-anatomic factors like temperature, electrode size and inter electrode distance. This study has been conducted to explore the bio heat transfer effect due to different atmospheric temperature for the assessment of EIM data. In this study, A 3D finite element model of human upper arm with rectangular shapes of electrode was developed for seven different ambient temperatures in the range of 20℃ to 50℃ to analyse temperature distribution inside biological tissue of human upper arm and consequently for the assessment of EIM parameters. We observed that there is no significant variation (p
Ahad, M. A. (2009). Electrical characteristics of rat skeletal muscle in immaturity, adulthood and after sciatic nerve injury, and their relation to muscle fiber size. Physiological Measurements, 1415. Ahad, M. A. (2010). Electrical properties of rat muscle after sciatic nerve injury: Impact on surface impedance measurements assessed via finite element analysis. Journal of Physics, 224(1), 012101. Ahad, M. A., & Rutkove, a. S. (2009). Electrical Impedance Myography at 50 kHz in the rat technique, reproducibility, and the effects of sciatic injury and recovery. Clinical Neurophysiology, 1534-1538. Chanmugam, A. H. (2012 ). Thermal analysis of cancerous breast model. ASME 2012 International Mechanical Engineering Congress and Exposition (pp. 135-143). American Society of Mechanical Engineers. Epstein, B. R. (1983). Anisotropy in the dielectric properties of skeletal· muscle·. Med Biol Eng Comput., 21, 51-55. Esper, G. J. (2006). Assessing neuromuscular disease with multifrequency electrical impedance myography. Muscle & Nerve, 34(5), 595-602. Farina, D. M.-N. (2004). Surface EMG crosstalk evaluated from experimental recordings and simulated signals Reflections on crosstalk interpretation, quantification and reduction. Methods Archive, 43, 30-35. Gabriel, C. G. (1996). The dielectric properties of biological tissues: I . Literature survey. Phys Med Biol., 41, 2231-2249. Grimnes, S. &. (2007). Sources of error· in tetrapolar impedance measurements on biomaterials and other ionic conductors. J Phys D: Appl Phys, 40(1), 9–14. Huang, H. W. (1994). Analytical solutions of Pennes bio-heat transfer equation with a blood vessel. Journal of Biomechanical Engineering, 116(2), 208-212. Jafarpoor, M. L. (2013). Optimizing electrode configuration for electrical impedance measurements of muscle via the finite element method. Biomedial Engineering, IEEE transactions, 60(5), 1446-1452. Jossinet, J. (1998). The impedivity of freshly excised human breast tissue. Physiol Meas, 19, 61-75. Kandala, S. K. (2013). Simulation of Discrete Blood Vessel Effects on the Thermal Signature of a Melanoma Lesion. ASME 2013 International Mechanical Engineering Congress and Exposition (pp. V03BT03A038-V03BT03A038). American Society of Mechanical Engineers. Kasraeian, S. A. (2010). A comparison of fine-needle aspiration, core biopsy, and surgical biopsy in the diagnosis of extremity soft tissue masses. Clinical Orthopaedics and Related Research®, 468, 2992-3002.(11), 2992-3002. Keyes, R. D. (1990). Nerve Conduction Studies and Electromyography. Canadian Family Physician, 36, 317. Martellosio, A. E.-L. (2017). Exposure limits and dielectric contrast for breast cancer tissues: Experimental results up to 50 GHz. In Antennas and Propagation (EUCAP), 11th European Conference on, (pp. 667-671). Narod, S. A. (2013). Are two-centimeter breast cancers large or small? Current oncology, 20(4), 205. Ortega-Palacios, R. L. (2010). Measurement of breast-tumor phantom dielectric properties for microwave breast cancer treatment evaluation. Electrical Engineering Computing Science and Automatic Control (CCE),2010 7th International Conference on , (p. 216=219). Rossmann, C. &. (2014). Review of temperature dependence of thermal properties, dielectric properties, and perfusion of biological tissues. Critical Reviews™ in Biomedical Engineering, 42(6). Rush, S. (1962). Methods of measuring the resistivities of anisotropic conducting media. J Res NBS, 66C, 217-222. Rutkove, S. B. (2009). Electrical impedance myography: background, current state, and future directions. Muscle & Nerve, 936-946. Sadiku, M. N. (2014). Elements of electromagnetics. Oxford university press. Shiffman, C. A. (1998). Angular dependence· of resistance in non-invasive electrical measurements of human muscle: the tensor model. Phys Med Biol., 43, 1317-1323. Tarulli, A. W. (2007). Multifrequency Electrical Impedance Myography In Amyotrophic Lateral Sclerosis. Graz, Austria: Springer Berlin Heidelberg. Wang, L. L. (2011). Assessment of alterations in the electrical impedance of muscle after experimental nerve injury via finite-element analysis. IEEE Transactions on Biomedical Engineering,, 58(6), 1585-1591. https://www.itis.ethz.ch/virtual-population/tissue-properties/database/dielectric-properties.
Research Data and Supplementary Material
Available for download on Friday, June 09, 2023